SPECIFICATIONS OF LAPAROSCOPIC SET Group -A ( Imaging and Insufflation devices ) Sno. Details of the instruments 1. 3 Chip Camera System, Color System, PAL comprising of (a) and (b) (a) Camera Control unit with integrated digital image processing module. It must be supplied with all standard accessories like mains cord, keyboard, connecting cable of length . 175 cms, BNC/ BNC video cable S video (Y/C) connecting cables, RGB connecting cable, DV cable. It must have following features Video outputs – Composite signal at BNC socket, SVideo Signal to 4 pol. Mini Din. Socket (2x), RGB Signal to 4x BNC sockets, DV Signal to 6 pin DV socket Inputs-Keyboard input for title generator and camera functions to 5 pin DIN socket Power Supply- 100-240 VAC, 50/60 Hz Certified to – IEC 601-1, 601-2-18, CSA 22.2 No. 601 L 2601 and CE according to MDD protection class 1/BF (b) Three chip camera head of color system, integrated, Par focal Zoom lens, f 25-50 mm (2x), with 2 freely programmable camera head buttons including plastic container for sterilization and storage. It must have following features : Image Sensor – 3 x ½ inch CCD chip Pixels – 752 (H) X 582 (V) pixels per chip (PAL) Resolution – min. 750 lines (Horizontal) Signal to Noise Ratio - > 60 dB AGC – Microprocessor controlled Min. Sensitivity – Parfocal Zoom, C mount 3 Lux (f=1.4 mm) Exposure control – 1/50 sec to 1/10000 sec (PAL) 2. 19 “ flat screen monitor, Desktop version, Color system, PAL, resolution max. 1280X1024 SDI, Composite, S-Video, RGB, DVI and S-XGA Input Brightness, 450 cd/m2, Contrast 650: 1 Power Supply, 100-240 VAC, 50/60 Hz. a. Cold Light Fountain XENON with one 300 watt , 15 V Xenon lamp Power supply 220-240 VAC, with color temperature upto 6000Kwith continuous manual adjustment of light intensity including mains cord. b. Spare Xenone bulb 300 watt Electronic CO2 insufflator with Gas flow of 19 litrs/ minute or more with touch keys for display of atleast 4 parameters (Insufflation, pressure, intra-abdominal pressure, gas flow and gas consumption) settings with optical and acoustic alarms for patient overpressure. It should have refilling safety system with release of pressure in case of over pressure and with automatic electronically controlled gas. To be supplied with main cord, silicon sterilizable tubing sets Universal wrench. Light Guide Cable – 4.8 to 5.0 mm dia, > 200 cm length,
Nos. 1 (ONE)
1 (One) 1 (ONE)
autoclavable, compatable with light source & telescope. Telescope : a. Telescope 30 cm in length, autoclavable, wide angle view, 10 mm dia, 0 degree forward viewing b. Telescope 30 cm in length, autoclavable, wide angle view, 10 mm dia, 30 degree forward – oblique viewing c. Telescope 30 cm in length, autoclavable, wide angle view, 5 mm dia, 30 degree forward – oblique viewing
(TWO) 2 (TWO) 1 (ONE) 1 (ONE)
Group -B ( Laparoscopic Instrumets ) Sno. Details of the instruments Nos. 1. Trocars and Cannulae a. Reusable, pyramidical tip trocars and cannula automatic and 5 manually operated, exchangeable functional door valve and guide (FIVE) tube, maintenance free stop cock for gas insufflation of size 11 mm b. Reusable pyramidical tip trocars and cannula automatic and 6 manually operated, exchangeable functional door valve and guide (SIX) tube, maintenance free stop cock for gas insufflation of size 6 mm c. Endotip cannula 11 mm of langth > 10 cm with multifunctional 1 valve. (ONE) 2. Veress Needle : a. Veress needle spring loaded of 130 mm length b. Veress needle spring loaded of 150 mm length a. Reducer sleeve for conversion of 10/ 11 mm to 5/6 mm b. Reducer 11/5 mm Laparoscopic hand instruments – a. Dissecting and Grasping Forceps autoclavable rotatable, dismantalable, insulated instruments with H-F connectors preferably with double action jaws of 5 mm diameter and 36 cm length with ergonomic handle (with or without rachet as per function). 1.Kelley’s dissecting and grasping forceps 2. Kelley’s dissecting and grasping forceps with long jaws 3. Right angled Kelley’s dissecting and grasping forceps 4. Grasping forceps, atraumatic fenestrated and curved with long jaws 5. Grasping forceps atraumatic with long jaws 6 Grasping forceps with fenestrated jaws 7. Grasping forceps with multiple teeth with jaws of 4.8 mm width 8. Grasping forceps with tiger jaws 2 X 4 teeth 9. Babcocks grasping forceps atraumatic fenestrated 10. Babcocks grasping forceps atraumatic jaws with multiple teeth fenestrated long 11. Claw forceps 2 X 3 teeth l.2 Bowel grasper fenestrated 13. Forceps for adenolysis b. Scissors : Autoclavable, rotatable, dismantalable, insulated instruments with H-F connectors preferably with double action jaws of 5 mm diameter and 36 cm length with ergonomic handle. 1. Metzenbaun Scissors 12 mm blade 2. Metzenbaun Scissors 17 mm blade 3. Blakesleys dissecting and biopsy forceps c. Instruments for cutting and coagulation (Unipolar/ Bipolar) 1. Dissecting spatula 5 mm 2. 2 Dissecting Hook cannula L shaped 3. 3 High frequency Unipolar cord 4. 4 Bipolar coagulation dismantalable forceps with facility of
2 (TWO) 1 (ONE) 2 (TWO) 2 (TWO)
2 (TWO) 1 (ONE) 1 (ONE) 2 (TWO) 2 (TWO) 2 (TWO 2 (TWO) 2 (TWO) 2 (TWO) 1 (ONE) 1 (ONE) 2 (TWO) 1 (ONE) 1 (ONE)
2 (TWO) 1 (ONE) 1 (ONE) 2 (TWO) 3(THREE) 5 (FIVE) 1 (ONE)
5. 6. 7.
changing, insert 5 Bipolar grasping dismantalable forceps with facility of changing, insert 6 Assorted Inserts for bipolar forceps 7 High frequency bipolar cord
1 (ONE) 4 (FOUR) 4 (FOUR)
Group -C (Clip Applicators, Needle Holder and Suturing Devices) d. Needle holders and suturing devices : 1. Needle holder ergonomic handle with ratchet left curved jaws with tungusten carbide inserts 5 mm diameter & length 33 cm 2. Assistant Needle holder Flamingo Jaw straight handle with ratchet size 5 mm length 33 cm 3. Fascial closure instrument 4. Integral knot pusher and thread catcher 5. GB speculam e. Clip applicator 1. Clip applicator for use with ethicon titanium clips (medium large) dismantling rotating with ratchet to lock the jaw part holding the clip size 10 mm length 36 cm 2. Clip applicator for use with ethicon titanium clips (medium) dismantling rotating with ratchet to lock the jaw part holding the clip size 10 mm length 36 cm. Group -D (Suction and Irrigation devices) f. Suction and irrigation devices 1 Cannula for puncture and aspiration 2 Suction/ irrigation tube with multile holes and two way stop cock 5 mm 3 Suction/ irrigation tube with multile holes and two way stop cock 10 mm 4 Suction irrigation pump – suction and irrigation pump, high flow system with 3 L / min capacity with all tubing for suction and irrigation cannula and jar for fluid of suction
1 (ONE) 1 (ONE) 1 (ONE) 1 (ONE) 1 (ONE)
2 (TWO) 2 (TWO) 1 (ONE) 1 (ONE)
Group -E (Miscellaneous/ Others) h. Miscellaneous 1 CO2 cylinder with high pressure hose – bulk type 2 CO2 cylinder with high pressure hose – 3 KG medical grade cylinder 3 CVT of 1 KVA 4. Trolley – mobile trolley for keeping laparoscopic equipments with sufficient number of power sockets
1 (ONE) 1 (ONE) 1 (ONE) 1 (ONE)
Note : Tenderers may quote any or all groups. Ernest money of groups tendered must accompany with tenders.
CHOLEDOCHOSCOPE SET It must comprise of following:
(A) CHOLEDOCHOSCOPE (2 mm channel) : -Fully immerssible in disinfectant solution. -Accurate observation backed up by increased image size. -Increased image guide fiber density. -Wide suction channel Field of View Working length Distal End dia. Insertion tube dia Depth of Field Bending Range : 120 degree or more (forward viewing) : 375 mm or more : 5.0 mm or less : 4.9 mm or less : 3 mm to 50 mm or better : Up – 160 degree Dn – 130 degree Channel dia. : 2.2 mm or more Biopsy forceps minimal visible distance : 5 mm or less from distal end. (B) SINGLE CHIP CCTV CAMERA : 10 bit digital signal processing with ¼ inch CCD chip. In built Moire filter for compatibility with fiberoptic endoscopes. Resolution: 470 horizontal lines. Signal to noise ration: More than 50 db Detachable and lockable adaptor with focusing facility. Video output: Y/c & Composite. Fully immerssible camera head & cable assembly. (C) COLOUR MONITOR: 14 inch medical grade LCD Monitor. (D) HALOGEN LIGHT SOUCE : One One One One
-It should be compact and light weight around 5 – 6 kg or less for easier transportability. -Should have autoclavable water bottle. -Should have 150 watts halogen lamp with stand by lamp option. -Should have Built in Air-pump for distention & automatic leakage testing. -Should be compatible with rigid and flexible endoscopes both, -Pump pressure 0.3 – 0.6 kg/cm2 (at Occ. / min.) 0.18 kg/cm2 or below (at 2000 cc/min.) (E) Accessories (a) Leakage tester (b) Compatible dormia wire basket (c) Compatible biopsy forceps (d) Choledeoscope grasping forceps 5mm (for use in Laparoscopy surgery)
One One Two One
Specification for Radio Frequency Ablation System The System o Should have Radio Frequency generator with feedback algorithm for continuous tissue impedance monitoring and automatic output adjustment. o Should have automatic current and impedance control feature. o Should have 200-watt power source with pulsating energy for generating larger coagulation volume. o Should have thermocouple to measure coagulation zone temperature and as the temperature cycle is complete electrode may be repositioned to measure coagulation zone temperature. o Should have automated treatment cycle to have successful typical tissue ablation in maximum 12 minutes. o Should have LCD status window to have user-friendly feedback during setup and operation. o Should have cooling electrode to eliminate tissue charring, reducing potential risk, and maximizing energy deposition. o Should have increased energy delivery results in larger ablation zones. o Should have multiple electrode length for use in percutaneous, Intraoperative and Laparoscopic applications. o The entire electrode can be imaged with CT, MRI, or Ultrasound guidance during the procedure. o Should have 17-gauge straight electrode with beveled tip for easy to insertion repositioning o Different electrode exposure lengths allow coagulation control for varying lesion sizes. o Should give electrodes of single or cluster type o Should have electrodes atraumatic in nature. and
EQUPIMENTS FOR ENDOSCOPY SUIT It must comprise of following:(A) UGI VIDEO-ENDOSCOPY SYSTEM.No- 1 *Slimmer/light weight and Fully Immerssible in disinfectant solution. *Compatible with variety of light sources (like Halogen / Xenon) *Three to Four remote switches on control body for maximum controls with user. *Compatible with semi-automatic leakage tester with its air flow & pressure regulation through compatible light source build in air pump. Field of view Depth of field : Distal end & Ins. Tube Dia. Channel dia. Bending range Working length : 145 deg or more (convertible to 120deg) 3mm to 100mm or better : 9.2 mm or less 2.8mm or more Up-210 deg Dn-90 deg L&R-100 deg. 1030mm or more
: : :
(B) COLONOSCOPE (VIDEO) No-1 *Slimmer/light weight and Fully Immerssible. *Three to Four remote switches on control body. *Graduated Stiffness for better operational control. *Compatible with semi-automatic leakage tester with its air flow & pressure regulation through compatible light source build in air pump. *Compatible with variety of light sources (like Halogen / Xenon) Field of view Depth of field Distal end dia. Insertion tube dia. Channel dia. Bending range Working length : : : : : : : (Wide)145 degree or more 3mm to 100mm or better 13.3mm or less 13.0mm or less 3.7mm or more Up & Dn-180 deg L&R-160 deg. 1675mm or more
(C) VIDEO DUODENOSCOPE (THERAPEUTIC) No-2 *Slimmer/light weight and Fully Immerssible in disinfectant solution. *Three to Four remote switches on control body. *Compatible with semi-automatic leakage tester with its air flow & pressure regulation through compatible light source build in air pump. *Compatible with variety of light sources (like Halogen / Xenon) Field of view Direction of view Depth of field Distal end dia. Insertion tube dia. Channel dia. Bending range Working length Minimal visible distance : : : : : : : : : 125 degree or more 5 deg. Backward Oblique 5mm to 60mm or better 13.2mm or less 12.5mm or less 4.2mm or more Dn. & left 90 deg. Up-120 & R-110deg. 1240 mm or more 10mm from distal end.
(D) R.G.B. O/P PROCESSOR & LIGHT SOURCE No-1 Processor / light source unit : Intergrated or separate unit with Xenon
or Halogen lamp (150~300 watts.) Processing : Digital signal processing. Image Display size/s : Full and small screen display on monitor Video out puts : RGB,Y/c & Composite output. Edge enhancement : Three steps Light control system : Automatic and manual control Cooling system : Forced air cooling. Spare lamp : Xenon or Halogen 150 – 250 watts. Lamp Change over system : Quick changeover system from main lamp to spare lamp and vice-versa Iris control, White balancing, “ON SCREEN” Colour Control (10-14 Incre. steps).
(E) ELECTRO SURGICAL UNIT : No-1 -Base frequency 350 kHz.& 1Mhz or more (for Spray) -Universal unit for Endoscopic & other Laparo surgical procedures. -Output Modes : Monopolar, Bipolar and Saline -Monopolar Cutting : Pure (300 W or more), Blend, URO (300 w or more) -Monopolar Coagulation : 120 W or more -Bipolar Cutting : Pure (90 W or more) -Bipolar Coagulation : Soft (two modes) and Hard -Preset Setting functions with CQM and audio / visual alarmings. -Foot switch and Patient plate (reusable). -Adaptor for compatibility with other make of cords. - Reusable patient plate and cord. ACCESSORIES and Others (General & Therapeutic): (a) Accessories Biopsy forceps (Standard UGI) Biopsy forceps(Aligator type Rat tooth) Biopsy forceps (Standard with needle) Two Prong haemoclip UGI (pre-loaded) set Two Prong haemoclip LGI (pre-loaded) set Polypectomy Snare (Oval Shape) Rat tooth foreign body forceps Two way bendable cannula Tripple lumen sphincterotome (3/20mm) Three step stone extraction balloon Autoclavable stone extraction basket (Hard) Lithotriptor comprising of a. Handle b. Metal sheath (Jacket) of Lithotriptor c. Basket of Lithotriptor d. Plastic sheath of Lithotriptor
(01 no) (1 no) (1 no) (2 nos) (1 no) (3 nos) (3 no) (1 no) (3 nos) (1 no) (2 nos) (1 no) (2 nos) (2 nos) (2 nos)
(b) LEAKAGE TESTER: No 1 Compatible with the above system with its air flow & pressure regulation through lightsource's air supply system.
(c) COLOUR MONITOR : Compatible RGB colour LCD monitor 14 inches. (d) ALMIRAH FOR ENDOSCOPE
(e) ENDOSCOPIC COUCH (f) CRASH CART (g) SUCTION MACHINE (h) ENDOSCOPIC TROLLEY (i) ON LINE UPS 1 KVA \ (with 30 min. backup)
(2 nos) (2 nos) (2 nos) (1 no) (1 no)
The unit should have the following essential features. 1. Generator and X-ray tube A) Generator should be microprocessor controlled with the following modes Power required – 2.5KW max i) Continuous fluoroscopy upto 15ma max with upto 30 f/s at 1024X1024 image matrix ii) pulsed fluoroscopy upto 23 mA max with upto 8 f/s iii) Digital Radiography mode The range of KV should be 40-110 KV for each mode. B) X-ray tube should have a i) Focal spot of nominal value suitable for fluoroscopy and radiography. ii) Nominal X-ray tube voltage 110 KV iii) Inherent filtration with details of esurient all thickness. iv) Automatic dose control C) Collimator Unit i) Iris diagphragm for radiation free collimation ii) Shutters / diaphragm for symmetric radiation free collimation and 360o rotation. iii) Indication for LIH. 2. C-arm Fully counterbalanced C-arm with integrated cables and electromagnetic brakes. Please give details of – (i) Angulation and orbital movement (ii) Horizontal movement (iii) Longitudinal movement (iv) Swivel range (v) Source to II distance (vi) Depth of immersion 3. A Image intensifier and X-ray TV system. i) Image intensifier should be at least 9” with zoom facility. ii) Input screen should be cesium iodide for excellent resolution and minimum noise. iii) Electron optics should allow consistent high resolution across the entire image field. Give details. iv) Grid : Give details B. The X-ray TV System It should be maintenance free with CCD technology. i) Image chain matrix at least 1K X 1K o ii) Digital image rotation + 360 4. Image Display i) 2 Nos. LCD monitors at least 18” in size. ii) Image matrix at least 1024 x 1024 5. Image acquisition storage and processing Must be a fully digital continuous imaging chain for acquisition, processing, storage, archiving and documentation. (i) Alphanumeric keyboard for entering patient data and for image annotation etc. (ii) Disk storage of minimum of 40,000 images in at least 1K X 1K matrix (iii). Image reversal, remask, invert etc. (iv) CD drive read/write . 7. Image Documentation (i) Should be possible to archive images on CD R/W format. (ii) Print images on paper / film 8. Essential Accessories (i) Online UPS of 4 KVA with at least 30 minutes back up for the C-arm. (ii) Lead aprons 4 nos. (iii) Thyroid shield 2 nos. (iv) Sterile covers for C-arm, X-ray tube and image intensifier 9. Kindly provide a copy of CE/FDA approval or quoted model.
quote DICOM connectivity for Laser Camera