VIEWS: 4 PAGES: 2 POSTED ON: 12/10/2012
Development of an Automated Solvent Extraction Apparatus Using Microchip for Heavy Element Chemistry H. Fujisawa, K. Ooe, Y. Komori, A. Kuriyama, R. Takayama, Y. Kikutani, H. Kikunaga, Y. Kasamatsu, T. Yoshimura, N. Takahashi, H. Habaa , Y. Kudoua , Y. Ezakia , A. Shinohara a Nishina Center for AcceleratorBased Science, RIKEN 1 Introduction Studies of chemical properties of superheavy elements and heavy actinides with atomic numbers ≥ 100 require rapid analytical techniques with repeatability because of the low production rates and short lifetimes of these nuclides. We have so far developed solvent extraction technique with a micro-chemical chip (MCC) and successfully applied to an on-line experiment. In this study, we have automated this apparatus and applied to a solvent-extraction experiment for nobelium. 2 Experimental Figure 1 shows a schematic diagram of the developed solvent extraction apparatus. Nuclear reaction products are transported to the chemistry laboratory with a He/KCl gas-jet transport system. The transported products are deposited on the collection site of a polychlorotriﬂuo- roethylene slider, are dissolved in aqueous solution, and are then subsequently fed into MCC (ICC-DY15, Institute of Microchemical Technology Co., Ltd.). An organic solution is fed from the other inlet of MCC. The aqueous and organic eﬄuents from MCC are separately collected on Ta discs. After evaporation to dryness, the samples are subjected to α spectrometry with Si detectors. The series of the operations is automated by using a LabView system (National Instruments Co.). Oﬀ-line tests of this apparatus were carried out with multitracers produced at the RIKEN Ring Cyclotron and single radiotracers produced at the RCNP AVF Cyclotron in Osaka Uni- versity. The radiotracers were stored in acetic acid/ammonium acetate solutions with pH of approximately 5.8. Batch experiments were performed as follows. 100 µL of the aqueous so- lution with various pH values and 100 µL of 0.01 M di(2-ethylhexyl) phosphoric acid-CCl4 solution was mixed in a test tube. The mixture was shaken for 10 min using a vortex mixer and was then centrifuged for 2 min. After phase separation, each phase was assayed for γ radioactivity. The oﬀ-line extractions with the apparatus were performed with the same solu- tions as those in batch experiments. Aqueous and organic solutions were fed into MCC with a ﬂow rate of 5 µL/min. Each phase from MCC was separately collected in test tubes and was assayed for γ radioactivity. 3 Results and Discussion The distribution ratio D is obtained by the following equation: D = Vaq Aorg /Vorg Aaq , where V shows volume and A is radioactivity. For MCC, V is replaced by the ﬂow rate. The log D values of strontium, which is a divalent cation as with nobelium, as a function of a pH value in the aqueous phase are shown in Fig. 2. The D values obtained with the apparatus were consistent with those by the batch experiments. The on-line extraction experiment of nobelium was performed using the apparatus described above. The 255 No (T1/2 = 3.1 min) isotope was produced in the 248 Cm(12 C, 5n) reaction at the RIKEN AVF Cyclotron. In this experiment, the AIDA system was used for α detection. The elapsed time from the dissolution of the reaction products to the start of measurement was approximately 4 min. As a result, we have obtained the D value of nobelium in the extrac- tion system although thorough corrections for such as cross-contamination between aqueous and organic phases are required. To perform chemical experiments of superheavy elements with atomic numbers ≥ 104, we will further improve the present apparatus for a more rapid chemistry. Microchip extraction Evaporation with IR heater Organic solution Aqueous solution Alpha detection Dissolution unit (AIDA) Collection Switching gas-jet line slider Transportation by He/KCl Gas-jet Gas exhaust beam Collection site Aqueous solution nuclear reaction He/KCl gas-jet He/KCl gas-jet products Figure 1: Schematic diagram of the automated solvent extraction apparatus. 1.5 1 logD 0.5 batch experiments apparatus tests 0 5.4 5.6 5.8 6 6.2 pH Figure 2: Plot of log D vs. pH for strontium with 0.01 M di(2-ethylhexyl) phosphoric acid. References  K. Ooe et al.: J. Nucl. Radiochem. Sci. 8, 59 (2007).  H. Haba et al.: J. Nucl. Radiochem. Sci. 3, 143 (2002).
"Development of an Automated Solvent Extraction Apparatus Using"