Cheat Sheet on
EAGLE Gas Monitors
1) Install Alkaline Batteries.
A) Unscrew the two large screws on the top of the case, then
carefully lift the top of the case and lay it aside.
B) Insert fresh batteries following the label on the inside of the
case. Place the top of the case in its original position, and then
secure it with the large screws you loosened in step A.
2) Starting Up the Eagle.
A) Connect the hydrophobic filter and probe tip to the sample
hose’s quick connect fitting.
B) Press and briefly hold down the POWER/ENTER button.
C) When the Eagle successfully completes its self check, the OK
message displays in place of the STAND BY message, then the
normal operating screen displays.
3) Fresh air zero needs to be done.
A) Take the Eagle to a non-hazardous location with fresh-air
B) Eagle need to run for one minute before zeroing.
C) Press and hold the AIR/s button until a tone sounds.
D) The Eagle automatically sets the combustible gas and toxics
circuits to zero and the oxygen circuit to 20.9%.
4) Eagle is ready for deployment.
5) Turning off the Eagle.
A) To turn off the Eagle, press and hold down the POWER/ENTER
button until GOOD-BYE displays, and then release the button.
6) Remove Alkaline Batteries.